Surgical Walking Stick
Our company's products are ISO11334-4 "single-arm operation of the walker requirements and test methods Part 4: three-legged or multi-foot cane" national standard as the reference standard for design and production, Q/DF4-2004 "Walking Aids" This enterprise standard as the implementation standard for design and production, Its structural features are as follows:
Main frame:The upper branch is made of aluminum alloy pipe material, the surface oxidation coloring treatment, pipe material specifications: diameter, wall thickness; 22mm1.2mm
Tripod:The tripod is composed of all-aluminum alloy welding, the surface is sprayed, and the foot tube material specifications: diameter 13mm, wall thickness; 1.2mm
Motherboard:The leg wrench is stamped into an arc by means of a mold, which is stronger, does not rust and is light in weight.
Foot pads:Four-foot landing structure, with rubber anti-slip foot pads, good landing performance, good stability, safe and reliable.
Performance:Height 8 gear adjustment, the product's inward stability performance is not less than 1.5 degrees, outward stability energy is not less than 4.0 degrees.